How to set the display inch, involving string pot CESI-S1000P and indicator HB961
The draw wire encoder you used is CESI-S1000P from CALT sensor, The resolution of CESI-S1000P is 0.1mm/pulse, that is 0.003937 inch per pulse, You can process the program according to such arithmetic logic.
Here we take our general HB961 indicator as an example. Refer to the accessory manual, The resolution of the cable encoder=A/(b*dot), that is, 1/(1*10)=0.1mm/pulse. If you want to convert to inch display, then by setting A=3937, b=10000, dot=—-.–, or A=3937, b=1000, dot=—.— , you can display in inches on the indicator.
In addition, the up button of HB961 has the function of resetting. Move the draw-wire encoder to the position you want, and then press the up key, the value displayed at the current position is ZERO.
Customized Loadcells Used In Sheet Metal Brake System
Company H is an American company that produces sheet metal brakes. It needs load cells to measure and control force during use. The customer originally
Customized Load Cell For Breeding Ingredient System
Company G is a Canadian company that provides intelligent equipment for poultry breeding companies. Its digital silo distribution system can monitor and control the feeding
Customized Hall Angle Sensor P3022
Company F is a Ukrainian company engaged in intelligent simulation seats. It needs angle sensors to control the steering angle and positioning. Previously, most of
Customize Your Own Draw-wire Position Sensor
Ultra-thin Type Protective Shell Additional Pulley Universal Pulley M12 Euro Connector Underwater Application